[FLAME WAVELENGTH ACCURACY TEST]
scan range                                    = 0.5
method1 name                                  = "flame 192.8"
method1 element                               = "Cu"
method2 name                                  = "flame 324.8"
method2 element                               = "Cu"
method3 name                                  = "flame 703.2"
method3 element                               = "Cu"
method4 name                                  = "flame 588.2"
method4 element                               = "Cu"
method5 name                                  = "fl As wvl accuracy"
method5 element                               = "As"
method6 name                                  = "fl Cu wvl accuracy"
method6 element                               = "Cu"
method7 name                                  = "fl K wvl accuracy"
method7 element                               = "K"
method8 name                                  = "fl Na wvl accuracy"
method8 element                               = "Na"


[FLAME BACKGROUND CORRECTION TEST]
%BCA                                          = 2.0
method1 name                                  = "bckgrnd test BC OFF"
method1 element                               = "Cu"
method2 name                                  = "bckgrnd test BC ON"
method2 element                               = "Cu"

[FLAME PHOTO NOISE BGC OFF TEST]
method name                                   = "BGC OFF 0 and 1 Abs"
method element                                = "Cu"
SD @ 0 abs                                    = 0.0007
SD @ 1 abs                                    = 0.0065
correction accuracy                           = 2

[FLAME PHOTO NOISE BGC ON TEST]
method name                                   = "BGC ON 0 Abs"
method element                                = "As"
photometric noise                             = 0.0026

[FLAME ABSORPTION TEST]
mean abs                                      = 0.55
%RSD                                          = 0.5
method name                                   = "SensitivityPrecision"
method element                                = "Cu"

[FLAME EMISSION TEST]
method1 name                                  = "Li Flame Em"
method1 element                               = "Li"
method2 name                                  = "Na Flame Em"
method2 element                               = "Na"
EHT                                           = 90

[GTA & PSD ANALYTICAL PERFORMANCE TEST]
mean abs                                      = 0.15
%RSD                                          = 4
rinse volume low                              = 1
rinse volume high                             = 2.7
R                                             = 0.996
method name                                   = "GTA Char Mass"
method element                                = "Cu"

[ZEEMAN WAVELENGTH ACCURACY TEST]
scan range                                    = 0.5
method1 name                                  = "zeeman 192.8"
method1 element                               = "Cu"
method2 name                                  = "zeeman 324.8"
method2 element                               = "Cu"
method3 name                                  = "zeeman 703.2"
method3 element                               = "Cu"
method4 name                                  = "zeeman 588.2"
method4 element                               = "Cu"
method5 name                                  = "ze As wvl accuracy"
method5 element                               = "As"
method6 name                                  = "ze Cu wvl accuracy"
method6 element                               = "Cu"
method7 name                                  = "ze K wvl accuracy"
method7 element                               = "K"
method8 name                                  = "ze Na wvl accuracy"
method8 element                               = "Na"

[ZEEMAN GTA TEST]
mean abs                                      = 0.15
%RSD                                          = 4
R                                             = 0.996
rinse volume low                              = 1
rinse volume high                             = 2.7
method name                                   = "Zeeman Char Mass"
method element                                = "Cu"

[ZEEMAN MSR TEST]
MSR low                                       = 70
MSR high                                      = 99
method1 name                                  = "MSR BC ON"
method1 element                               = "Cu"
method2 name                                  = "MSR BC OFF"
method2 element                               = "Cu"

[ZEEMAN BACKGROUND CORRECTION TEST]
method1 name                                  = "BCA BGC off"
method1 element                               = "Au"
method2 name                                  = "BCA BGC on"
method2 element                               = "Au"
BCA                                           = 3.8

[SIPS DILUTION ACCURACY TEST]
method1 name                                  = "SIPS sps2"
method2 name                                  = "SIPS sps3"
method3 name                                  = "SIPS sps5"
method4 name                                  = "SIPS manual"
method element                                = "Cu"
dilution accuracy low                         = 96
dilution accuracy high                        = 104
%RSD                                          = 3

[VGA GAS AND LIQUID FLOW TEST]
sample liquid flow low                        = 6
sample liquid flow high                       = 9
acid liquid flow low                          = 1
acid liquid flow high                         = 2
reductant liquid flow low                     = 1
reductant liquid flow high                    = 2
inert gasA flow low                           = 40
inert gasA flow high                          = 45
inert gasB flow low                           = 40
inert gasB flow high                          = 45

[SPS DILUTION ACCURACY TEST]
method1 name                                  = "SPS3 Dilutor Test"
method2 name                                  = "SPS5 Dilutor Test"
method element                                = "Cu"
dilution accuracy low                         = 96
dilution accuracy high                        = 104

[ETC 60 TEST]
method name                                   = "ETC-60 operation"
method element                                = "As"

