Mass Annotation Dialog Box

Use this dialog to specify which UV and MS chromatograms are to be annotated

with the value of the most abundant mass in the MS spectra corresponding to

each peak, and to set the style of the annotation.

Note: This macro does not work on SIM data, only scan data.

Annotate Group Box

Select the chromatograms to be annotated:

(TIC) Total Ion Chromatogram

(EIC) Extracted Ion Chromatogram (Note: All EICs will be annotated.)

(BPC) Base Peak Chromatogram

 	(UV)  UV Chromatograms (Note: All UVCs will be annotated.)

      	UV Delay - time in minutes of delay between UV and MSD

detectors. Default: 0.0 minutes. Range: 0.0 to 9.999 minutes.

Offset Group Box

Parameters to adjust vertical position of peak

      annotations to keep annotations of large peaks on screen.

Median % - limit, as a percentage of the full spectra height, at which

      peak annotation shift changes direction.  Annotations of peaks

      whose height are below this limit are shifted above their normal

      position using the % Up setting.  Annotations of peaks whose

      height are above this limit are shifted below their normal

      position using the % Down setting. Default: 60%. Range: 0 - 100%.

% Up - Default: 0%.  Range: 0 - 100%.

% Down - Default: 60%.  Range: 0 - 100%.

Style Group Box

Set the appearance and position of the annotations.

Extra @ ??? Degrees

The mass annotation is offset from the retention time by the specified

number of degrees. Default: 315 degrees.

Range: 0 to 360.

Append To Ret Time

The mass annotation is appended to the existing retention time.  This

is the default selection.

Replace Ret Time

The mass annotation replaces the existing retention time.

Number of Ions

The annotation lists the n th largest ions from the MS spectra. Default: 1.

Range: 1 - 3.

Text Color

Determines the annotation color. Default: red. Range red, blue, green,

black.